Lucky for you, the CytoSMART® Omni-FL requires minimal cleaning for reliable operation. Unplug the USB
and power cables before cleaning the device. Use a damp, lint free wipe to clean the outer case with
isopropyl alcohol (IPA) or (70%) ethanol.
If the optical window is dirty or contains dust, clean it using a soft, lint free wipe. You can use IPA or
ethanol in case the optical window needs to be cleaned thoroughly. Do not use abrasive chemicals or
rough materials to clean, as this may scratch the optical window. A scratched optical window can be
repaired but this is not included within the product’s warranty.
Important: Do not use acetone to clean the CytoSMART® Omni-FL . Also, do not autoclave the CytoSMART® Omni-FL , this will result in damage to the electronics.
The CytoSMART® Omni-FL does not need regular maintenance. Do not perform any repairs or service on the device yourself. Contact CytoSMART® if a problem should arise, via firstname.lastname@example.org.